Detecto WINDDE Dynamic Data Exchange Weight Capture to PC Software
Gift-wrapping is not available for this product.
Features: Detecto WINDDE Dynamic Data Exchange Weight Capture to PC Software Weight indicators may be connected using standard computer serial ports or via ethernet using TCP/IP. WinDDE Server communicates with up to 20 weight indicators attached to a computer. Dynamic Data Exchange Program for transferring weight data to Windows based programs on your PC from 758C digital the capture feature may be based on the indicator print button to capture a row of data when the print key is pressed on the indicator. Product Information:
Detecto WINDDE Dynamic Data Exchange Weight Capture to PC Software,
The weight and status of the weight indicators is made available to spreadsheets, databases, and industrial control applications by means of Dynamic Data Exchange, or NetDDE for networked computers. Weight and time stamp information may be automatically captured to Excel creating a new row of information for each stable weight above threshold. Industrial Control Software such as Rockwell RSView, Automation Direct, Lookout Direct, and spreadsheet programs, such as Microsoft Excel may be used. A spreadsheet cell, such as A1, shows continuously updating weight. The spreadsheet may show a continuously updating bar graph. Formulas in the spreadsheet show a flashing indication if the weight exceeds or drops below certain levels. After a weight is placed on the scale and the indicator sees a stable weight, the weight is placed in a spreadsheet cell, such as A5. The operator removes the weight, and the procedure is repeated, but the weight value is placed in the next spreadsheet row, such as A6. After a weight is placed on the scale, the operator at the indicator may press the Print key on the indicator. Instead of printing, the printout information from the indicator is sent to the computer. The computer sets the spreadsheet cells, such as E3 Gross Weight, F3 Tare Weight, G3 Net Weight, etc. The operator then may remove the weight, and the procedure is repeated, but information is placed in the next spreadsheet row, such as E4, F4, G4, etc. After a weight is placed on the scale, the operator may type information at the computer, such as an ID number to associate the weight. The computer stores the information in a database or spreadsheet. The operator then may remove the weight. SPECIAL ORDER; No returns. Condition: New Shipping Weight: 2lb
This item will be shipped from the manufacturer and is not one we carry in stock. Please contact us for availability, as we cannot always guarantee order fulfillment.